This simple, low cost, battery operated unit supplies the polarising voltage for a single electrode and produces an analogue output for a chart recorder.
The controls allow the polarising voltage (which is displayed on a meter) to be adjusted. The recorder output voltage is adjusted by a five position, coarse adjustment switch and a ten-turn fine potentiometer which enables the electrode to be calibrated to the recorder. The electrode and the recorder are connected via DIN sockets on the front panel. A stirrer is not included.
|Polarising Voltage||0–1.5 V|
|Recorder Output||0–30 mV|
|Power Supply||1 × C type battery (supplied)|
|Size||200 mm (width) × 120 mm (depth) × 70 mm (height)|
Rank Brothers Ltd reserves the right to change specifications in the light of continuing development.
These operating manuals are in PDF format. They can be opened in the free Adobe Acrobat Reader software.
For a working oxygen measurement system you will need a controller and an electrode (the Polarising Circuit also requires a separate stirrer). Rank Brothers offers a range of Perspex oxygen electrodes and glass oxygen electrodes suitable for use with this controller.
|OXY031A||Polarising Circuit (stirrer not included)|